EmbaGYN™ Pelvic Floor Exerciser is designed with state-of-the art electronic features including:
The strength of the stimulation is under your control. You can gradually increase or decrease the intensity whenever you want to make sure the stimulation always feels comfortable.
The EmbaGYN has intelligent memory that enables the unit to remember the last settings you used,and return to similar settings at your next use.
EmbaGYN also has Usage Memory that records the length of each session and the strength level you used.This can help you keep track of your progress.
To activate hold down the Time Selector and strength UP buttons together for 3 seconds. The screen will show the time in minutes (to the nearest hour) and strength in rnA. Hold down Time Selector and strength DOWN button together for 3 seconds to reset the memory to zero.
This feature automatically turns the device off after 20 minutes of use to protect the pelvic floor muscles from being overworked. If your physician recommends sessions that are longer than 20 minutes,you can adjust the device to override the safety shutoff and allow a longer session.
EmbaGYN monitors the contact between you, the Trainer, and the EmbaGYN Control Unit. If either of these go outside of a standard range, while the strength is above 20.0, the unit will flash "LEADS", beep three times, and return strength to zero.
The most common causes are: